CT Lab HX
Flexible geometry for 2.1 um/voxel resolution Flexible geometry for 200 mm FOV Versatile and compact benchtop design
The versatile and compact micro CT scanner
What is CT Lab HX?
Rigaku CT Lab HX is a benchtop micro CT (computed tomography) scanner. The adjustable SOD (source-to-object distance) and SDD (source-to-detector distance) make this benchtop micro CT scanner flexible. It covers from 2.1 um voxel resolution in the high-resolution mode and 200 mm FOV (field of view) in the large FOV mode. The CT Lab HX is equipped with a 130 kV - 39 W high power X-ray source. The X-ray source settings and X-ray filters are adjustable to optimize the X-ray energy to various sample materials and sizes.
CT Lab HX features:
- Flexible design to cover high-resolution to large FOV
- 2.1 um voxel resolution
- 200 mm FOV
How do I achieve high resolution?
Rigaku CT Lab HX uses the cone-beam geometry. This geometry uses the X-ray beam divergence to magnify the sample image. The magnification factor is determined by the ratio of SOD (source-to-object distance) and SDD (source-to-detector distance). So to achieve high resolution, you can increase the SDD and/or decrease SOD. CT Lab HX has the flexibility to do them both in its compact design.
With the long SDD and short SOD settings, you can achieve 2.1 um voxel resolution (~ 5 um spatial resolution) on the CT Lab HX. This resolution can reveal micro cracks, voids, fibers in composites, etc.
How do I achieve 200 mm FOV?
The FOV (field of view) you can cover in one scan is limited by the size of the detector. However, the offset scan mode doubles the effective detector width to cover a wide FOV without having to change the detector. Rigaku CT Lab HX utilizes this scan mode to achieve 200 mm FOV without having to change the detector or compromising its compact system size.
The 200 mm FOV enables a scan of an entire smartphone, 3D printed objects, fruits, etc.
Do I need a large CT scanner to cover both high-resolution and large FOV?
No, you don't need a large floor standing type CT scanner to cover a wide range of resolution and FOV for most samples under 200 mm in size and made of elements lighter than aluminum.
Rigaku CT Lab HX has the flexibility to change both SOD (sample-to-object distance) and SDD (source-to-detector distance) to cover from 2.1 um to about 100 um voxel resolution. And the offset scan mode can cover 200 mm FOV.
This flexible and compact design makes CT Lab HX an affordable research-grade micro CT scanner ideal for materials science imaging centers.
CT Lab HX Specifications
|2.1 — 100 um
|Field of view (FOV)
|5 — 200 mm
|Maximum sample size
|200 mm diameter x 270 mm height
|Speed (shortest scan time)
|Cone beam geometry
|39 W traditional micro source
|W anode, adjustable applied voltage 30 - 130 kV
|Detector pixel size
|2352 x 2944 pixels
|980 (W) x 580 (H) x 700 (D) mm (PC not included)
|Approx. 380 kg
How can we help you?
We are here to help you
If you are looking for more information about the CT Lab HX micro CT scanner, such as a quotation, installation requirements, etc., please fill out the form and we will get that for you.
If you just have a general question or want to talk to someone, click the button below to book a meeting with one of our experts.
Rapid-release pain medication capsule comparison
Multivitamin tablet analysis
Compression methods comparison
Particleboard packing density analysis
Pretzel coating analysis
Processed meat fat and protein analysis
Breading air oil distribution analysis
Sandstone grain size analysis
Sandstone pore network analysis
Application Note & Video
Ammonite fossil scan
Carbonate porosity analysis segmentation comparison
Granite phase fraction analysis
Rock crack analysis
Sand oil and brine segmentation
Sandstone porosity comparison
Bottlecap air gap analysis
Golf ball layer thickness analysis
Nesting doll dimensional analysis
Plastic parts quality control and reverse engineering
Water flosser tip stress simulation
3D printed plastic dimensional analysis
Aluminum die casting void analysis
Aluminum part dimensional analysis
Application Note & Video
Tomato plant root system analysis
Broken bonding wires in LED chip
Delamination in lithium-ion batteries
Solder Voids in PCB
Plastic Locking Parts Dimensional Comparison
Coral Surface Area and Porosity Analysis